Loop-Mediated Isothermal Amplification (LAMP)
Loop-mediated isothermal amplification (LAMP) is a sensitive molecular tool suitable for use as a near point-of-care test for the diagnosis of malaria.
Loop-mediated isothermal amplification (LAMP) is a sensitive molecular tool suitable for use as a near point-of-care test for the diagnosis of malaria in low resource settings . The LAMP methodology—first published in year 2000—relies on isothermal deoxyribonucleic acid (DNA) amplification employing the Bacillus stearothermophilus (Bst) DNA polymerase with strand displacement activity . LAMP can therefore be performed at a single temperature with a simple heating block or water bath, reducing the need for sensitive and expensive machinery such as polymerase chain reaction (PCR) thermocyclers. The LAMP reaction is primed by a specific set of four to six primers that identify distinct regions on the target DNA. The design of the primers results in DNA loop formations and several inverted repeats of the target DNA . This autocycling strand-displacement DNA synthesis makes the amplification highly efficient and specific, allowing the amplification of a few DNA copies to 109 copies under isothermal conditions in less than one hour, reducing the time-to-result . DNA amplification can be detected by eye by a change in turbidity caused by white precipitate of magnesium pyrophosphate formed during the reaction, or under ultraviolet (UV) fluorescence if a fluorescent indicator such as calcein is added to the reagents . Visual detection avoids the need for opening the reaction tube post-amplification, hence the reaction is conducted in a closed system which reduces the risk of DNA contamination . Furthermore, the Bst polymerase is more robust towards inhibition than Taq polymerase in conventional PCR, making it suitable for use with simple and field friendly DNA extraction methods, but maintaining a sensitivity comparable to PCR .
The first malaria specific LAMP assay targeting the Plasmodium falciparum 18S ribosomal ribonucleic acid (rRNA) genes was published in 2006 . This was followed by Plasmodium genus and species-specific LAMP assays published in 2007 . Since then, over 40 different LAMP methods have been developed. These developments—some of which were recently reviewed —have aimed at improving the sensitivity of the assay by targeting mitochondrial DNA  or alternative gene targets . Other methods have aimed at improving or mitigating DNA extraction processes , or improving the read-out of the results by incorporating different dyes  or by combining LAMP with lateral flow dipsticks providing a similar result format to malaria rapid diagnostic tests (RDTs) .
Recent meta-analyses evaluating the diagnostic accuracy of LAMP for malaria have detailed high sensitivity and specificity of LAMP when compared to microscopy, PCR, and RDTs in both endemic and non-endemic settings (Table 1). The pooled sensitivity and specificity of LAMP has largely remained greater than 95% whichever the comparator , with an area under the curve of greater than 0.98 demonstrating that malaria LAMP is a test with excellent diagnostic performance . These meta-analyses concluded that the LAMP method is a robust tool for diagnosing malaria in both symptomatic and asymptomatic individuals , and that LAMP is one of the most promising new diagnostic tools for use in malaria endemic settings .
Table 1. Pooled estimates of the diagnostic accuracy of malaria loop-mediated isothermal amplification as determined in three meta-analyses.
(%; 95% CI)
(%; 95% CI)
|LAMP vs. LM||98; 94–99||97; 85–99||ND|||
|LAMP vs. PCR||96; 79–99||91; 68–98||ND|||
|LAMP vs. LM||97; 96–98||96; 94–97||0.98|||
|LAMP vs. RDT||97; 92–99||96; 92–98||0.98|||
|LAMP vs. PCR||97; 96–98||96; 94–97||0.98|||
|Pv LAMP vs. PCR||95; 80–99||96; 86–99||0.98|||
|Pan LAMP vs. PCR||95; 91–97||98; 95–99||0.99|||
|Pf LAMP vs. PCR||96; 94–98||99, 96–100||0.99|||
|Pv LAMP vs. PCR||98; 92–99||99, 72–100||1.00|||
95% CI: 95% confidence interval; AUC: area under the curve; LAMP = loop-mediated isothermal amplification; LM = light microscopy; PCR = polymerase chain reaction; RDT = rapid diagnostic test; Pv = P. vivax; Pan = Pan-Plasmodium; Pf = P. falciparum; ND = not determined.
2. Commercially Available LAMP Kits
Fifty-eight of the 148 identified malaria LAMP publications employed one of two brands of field-stable and CE-marked (where CE stands for Conformité Européenne” in French or “European Conformity” in English) malaria LAMP kits that are now commercially available: the Illumigene® (now Alethia ®) malaria LAMP (Meridian Bioscience Inc., Cincinnati, OH, USA) and the LoopampTM Malaria Detection Kits (Eiken Chemical Co., Tokyo, Japan). Neither of the kits require a cold chain, as they contain ready to use lyophilized reagents that are stable at room temperature . Both brands target the mitochondrial DNA of Plasmodium species for genus-level identification. In addition, the Loopamp Malaria Pf Detection Kit and the more recent Loopamp Malaria Pv Detection Kit can differentiate P. falciparum and P. vivax infections from other species, respectively (Table 2) .
Table 2. Comparison between the Illumigene® (Alethia®) malaria LAMP (Meridian Bioscience Inc.) and the LoopampTM Malaria Pan Detection kits (Eiken Chemical Co.).
|Species identification||Pan-Plasmodium||Pan-Plasmodium; P. falciparum; P. vivax 1|
|Sample type||Fresh/frozen blood||Fresh/frozen blood or dried blood spots|
|Methods of DNA extraction||Illumigene malaria; Illumigene malaria PLUS||PURE DNA Extraction Kit; Boil and spin; other methods 2|
|Limit of detection 3||2.0 p/µL for P. falciparum and 0.1 p/µL for P. vivax||1–2 p/µL|
|Required equipment||Illumipro-10™ incubator||Centrifuge, heat block/water bath, UV light, (or LA-500 turbidimeter/HumaLoop M)|
|Number of samples per run||10||16, 46 or 94 4|
|Read out of results||Turbidity in Illumipro-10™ incubator or by eye||Turbidity in turbidimeter or by eye; fluorescence under UV light|
|Primary area of use||Malaria diagnosis in non-endemic settings||Malaria prevalence surveys in endemic settings|
|Cost per test 5||28 EUR||5.2 EUR|
1 Three separate kits. 2 Including Chelex extraction, column-based extraction methods, and a high-throughput DNA extraction platform. 3 Under laboratory conditions. 4 If used together with a turbidimeter or HumaLoop M platform, regular heating block, or high-through put extraction setup, respectively. 5 Does not include cost for equipment.
Both the Illumigene and the Loopamp kits can be performed on fresh or frozen blood, in combination with simple and quick DNA extraction methods. The Loopamp kits can also be performed on dried blood spots . Both brands provide high analytical sensitivity, with a time-to-result of 1–2 h . Both kits are easy to handle but require some basic laboratory skills as they require blood sample preparation and measures to avoid DNA cross-contamination . The risk for DNA contamination is however minimized, as both kits utilize a robust closed system, where tubes with amplified products are never opened (unlike conventional or nested PCR) . A drawback of the LAMP methodology is that it does not provide parasite quantification.
2.1. The Illumigene® (Alethia®) Malaria LAMP
The Illumigene Malaria LAMP kit is easy to perform, with DNA extraction and result readout included in the same kit. DNA extraction is conducted by a simple gravitation system that employs lysis buffers and columns to simplify sample preparation. In this approach, whole blood is mixed with a lysis buffer and either used directly in the LAMP assay (simple filtration assay) or passed through a column that purifies the DNA via gravity (Illumigene malaria PLUS) . Both procedures rely on chemical lysis and produce amplifiable DNA within 10 minutes, requiring only the use of micropipettes, DNase/Rnase free sterile pipette tips, and latex gloves. Each test device consists of two tubes, a test tube with primers targeting the Plasmodium genus and a control tube with primers to detect a housekeeping human gene used as a DNA extraction and amplification control.
Result readout with an Illumigene Malaria Illumipro-10™ incubator/reader measures turbidity induced by the formation of magnesium pyrophosphate. Ten samples can be analyzed per run in this platform . The detection limit of the assay is 2.0 parasites per microliter (p/µL) or 0.3 p/µL for P. falciparum (depending on which extraction method is used) and 0.1 p/µL for P. vivax . A disadvantage with the kit is that no species differentiation is possible, and there are a few reports of occurrence of invalid results .
The Illumigene kit is preferably conducted on venous EDTA whole blood samples, making it possible to perform the LAMP assay with some delay. This blood sampling method can however be challenging in resource limiting settings . Furthermore, the Illumigene kit has a high cost (~28 euros per sample plus the cost of the Illumipro-10™ incubator/reader), limiting its use to developed countries for diagnosis of imported malaria .
2.2. The LoopampTM Malaria Detection Kits
The Loopamp Malaria Detection Kits have been endorsed by the Foundation for Innovative New Diagnostics (FIND) who has developed standardized procedures for their use . The Loopamp Malaria Pan Detection Kit is typically used for malaria screening with an analytical sensitivity of 1–2 p/µL . Positive samples can thereafter be retested using the Pf or Pv detection kits for further species identification .
DNA can be extracted with the commercially available Loopamp PURE DNA Extraction Kit, consisting of a series of interlocking plastic components providing a closed system for preparation of a blood aliquot requiring only a heat block or incubator, micropipettes, DNase/Rnase free sterile pipette tips, and latex gloves . The Loopamp kits can also be used with a cheaper, quick and easy, boil and spin DNA extraction method, but requires additional equipment such as a centrifuge . The LAMP reaction can either be conducted in a simple heat block or water bath, or in a real-time turbidimeter. The formation of magnesium pyrophosphate resulting in turbidity, or fluorescence produced by the release of calcein upon amplification in positive samples, can immediately be read by eye or by using a simple UV light source, respectively . However, the use of a UV-lamp for reading results is comparably less objective than a turbidimeter .
The LA-500 real-time turbidimeter (Eiken Chemical Co.) or the HumaLoop M (Human diagnostics Worldwide, Wiesbaden, Germany) platform for sample preparation, amplification, and easy visual result reading under a built in UV light can hold 16 tests per run. If a simple heating block system is used then batches of up to 46 samples can be run together with a positive and negative control (the maximum number of tubes that fit on a regular heat-block), making it a useful tool for prevalence surveys in endemic areas . Furthermore, a high throughput DNA extraction platform that can extract up to 94 samples at a time has been assessed, albeit with varying performance . The Loopamp kits cost about 5.20 euros per test, excluding the costs for DNA extraction and equipment.
3. Performance and Application of Commercial LAMP Kits in Malaria Endemic Settings
The performance of LAMP has been evaluated in malaria endemic field settings in over 30 publications, of which 15 studies employed the LoopampTM MALARIA Detection kits and two studies the Illumigene® (Alethia) malaria LAMP. Overall, LAMP has shown ≥95% pooled sensitivity and specificity for detecting both P. falciparum and P. vivax infections when used in endemic settings . The consensus among studies that have employed one of the commercial kits in malaria endemic settings is that the methods are easy to perform after only 3–5 days laboratory training , even in the remotest of field settings .
3.1. Performance of LAMP in Asymptomatic and Low-Density Infections
As malaria transmission declines in areas of successful malaria control, the relative proportion of asymptomatic and low-density malaria infections increases . These infections are often subpatent, i.e., falling below the detection limit of conventional malaria diagnostic tools such as microscopy or RDTs. Hence, as malaria transmission decreases, new and more sensitive field applicable screening tools are needed for the detection and management of very low-density infections, especially if malaria elimination is to be considered possible .
The improved analytical sensitivity of LAMP, resulting in the detection of significantly more infections compared to RDT or microscopy , and the reduced time-to-result compared to PCR , has put LAMP forward as a promising tool for near point-of-care detection of low parasite density infections in asymptomatic carriers, especially in low endemic and pre-elimination areas . However, despite its improved analytical sensitivity, LAMP may still miss a substantial proportion of the asymptomatic reservoir of very low-density infections , especially in low transmission settings (Table 3). Although LAMP may provide a better understanding of the prevalence and distribution of low-density asymptomatic infections than conventional diagnostic tools, a rapid turn-around time may not necessarily be the highest priority in malaria epidemiological surveys where highly sensitive PCR-based methods are still often the method of choice . On the other hand, results should be made available within 48 hours of testing in detect-and-treat approaches such as reactive case detection (i.e., screening and treatment of household members and neighbors of passively identified index cases), and mass or focal testing and treatment strategies. Further studies are however needed to assess if employing LAMP will significantly improve the impact such resource-intensive elimination strategies have on malaria transmission .
Table 3. Performance of malaria loop-mediated isothermal amplification for detecting asymptomatic low-density infections.
|Setting||Sample Size 1||Sample Type 2||LAMP Method||Comparator||Prevalence (%)||Mean Parasite Densities
(%; 95% CI)
(%; 95% CI)
|Zanzibar, Pre-elimination||996||Fresh blood + Boil and Spin||Loopamp Pan/Pf||RDT;
|RDT: 1.0 LAMP/PCR: 1.8||26 (range: 0–4626).||Pan LAMP: 83.3; 59–96
RDT: 55.6; 31–79
|Pan-LAMP: 99.7; 99–100
RDT: 100; 99.6–100
|Zanzibar, Pre-elimination||3983||Fresh blood + Boil and Spin||Loopamp Pan/Pf||RDT;
|RDT: 0.5; LAMP 1.6||ND; 71% of LAMP positives <LOD of RDT||RDT: 24.6; 15–37||RDT: 99.9; 99.7–100|||
|Zanzibar, Pre-elimination||3008||Filter device +HTP extraction||Loopamp Pan/Pf||RDT;
|RDT: 0.4; qPCR: 1.6; HTP-LAMP 0.7||1.8 (range: 0.1–770)||HTP-LAMP: 40.8; 27–56
Chelex-LAMP: 49; 34–64
HTP_LAMP >2: 54; 25–81
HTP_LAMP ≤2: 36; 21–54
|Eswatini (formally Swaziland), Very low transmission||10890||DBS + Chelex extraction||Loopamp Pan/Pf||RDT Ref; LAMP
LAMP Ref: nPCR
|RDT: 0.6; LAMP: 1.7||ND; 67% of LAMP positives <LOD of RDT||LAMP: 72.2; 63−80
RDT: 33.4; 33–35
|LAMP: 98.0; 97−98
|Namibia, Low transmission||2642||DBS and used RDTs + Chelex extraction||Loopamp Pan||RDT Ref: nPCR||RDT: 0.9; LAMP 1.8||ND; 51% of LAMP positives <LOD of RDT||LAMP on RDT: 95.4; 84–99
LAMP on DBS: 95.5; 85–99
RDT: 9.3; 2.6–22
|All > 99|||
|Colombia, Varied transmission||980||Fresh blood + Boil and Spin||Loopamp Pan/Pf||LM Ref: qPCR||LM: 0.2; LAMP: 6.6; qPCR: 7.2||ND, (range: 1–897)||Pv: 90.9; 80–97
Pf: 100; 78–100
|All > 99|||
|Peruvian Amazon, Low to moderate transmission||1167||Fresh blood + Boil and Spin||Loopamp Pan/Pf||LM Ref: qPCR||LM: 4.9; LAMP: 21.9||10 (CI95% 7.5–13)||LAMP: 91.8; 88–95
LM: 20.3; 16–26
LM: 98.0; 95–99
|Uganda, High transmission||554||DBS + Chelex extraction||Loopamp Pan||LM Ref: qPCR||LM: 18.2; LAMP: 37.2; qPCR: 48.9||LAMP neg:
0.1 (CI95% 0.07–0.2)
5.7 (CI95% 3.0–10.8)
|LAMP: All LM negs: 44.7
≥0.01–<0.1 p/µL: 10.8
≥0.1–<1 p/µL: 40.9≥1 p/µL: 81.5
|All LM negs: 94.0|||
1 All asymptomatic; 2 All samples from finger prick; p/µL = parasite per microliter; 95% CI = 95% confidence interval; Ref = reference method; Pf = P. falciparum; Pan = Pan-Plasmodium; RDT = rapid diagnostic test; LM = light microscopy; qPCR = quantitative PCR; nPCR = nested PCR; LAMP = loop-mediated isothermal amplification; DBS = dried blood spot; HTP = high through put; <LOD = below the limit of detection; ND = not determined.
3.2. Application of LAMP in Prevalence Surveys in Malaria Endemic Settings
Eight studies—all of which were conducted in sub-Saharan Africa—have applied commercial LAMP kits as a screening tool in malaria prevalence surveys. For instance, LAMP has been used in several studies in Uganda to investigate the effect of HIV infection on malaria incidence ; the effect of indoor residual spraying on the prevalence of asymptomatic infections ; for assessing the prevalence of microscopic and submicroscopic malaria infections in different transmission sites ; and the prevalence of submicroscopic infections among schoolchildren . In these studies, dry blood spots on filter paper were collected at point-of-care. DNA extraction was conducted by a Chelex-based method, followed by the Loopamp Malaria Pan Detection Kit. Depending on age, prevalence of asymptomatic infections, and malaria transmission intensity, the LAMP positivity rate was two to ten times greater than the positivity rate by microscopy or RDT .
LAMP has also been applied for assessing risk factors and spatial clustering of asymptomatic malaria though active and reactive surveillance in low transmission settings . In these studies, dry blood spots were collected for later Chelex extraction followed by screening with the Loopamp Pan Detection Kit. LAMP detected significantly more infections than RDTs, with clustering of asymptomatic low-density infections around index cases . In the study by Smith et al. RDT identified only 17% of the index and neighbor cases detected by LAMP, suggesting that infections missed by RDT during reactive case detection may be responsible for 50–71% of transmission from humans to mosquitoes .
A single study performed in Congo has applied the Illumigene Malaria kit. Venous blood (4 mL) with EDTA was taken from 1088 children aged between 6 and 59 months for assessing the prevalence of anemia and its relationship with asymptomatic submicroscopic Plasmodium infection. Malaria prevalence was 16% and 34% by microscopy and LAMP, respectively and submicroscopic Plasmodium infection was found in 22% of the children .
4. Pros and Cons of Commercially Available LAMP Kits in Low Resource Settings
It is unlikely that LAMP will replace conventional diagnostic tools such as RDTs and microscopy in the point-of-care diagnosis of clinically symptomatic malaria in endemic settings . LAMP in its current formats does not completely fulfil the World Health Organization (WHO) ASSURED criteria (i.e., being Affordable, Sensitive, Specific, User-friendly, Robust and rapid, Equipment free, and Deliverable) for identifying appropriate diagnostic tests for resource-constraint settings, and some improvements are likely to be necessary for this tool to become readily available . Whilst the commercially available diagnostic kits certainly offer user-friendliness, with high specificity and robust, rapid, and deliverable assays that are independent of a cold chain, they are not equipment free and require a reliable source of electricity. Despite the ease of use, LAMP still requires good enough facilities for preparing basic molecular assays, ideally with separate workstations for avoiding DNA-cross-contamination . In addition, the technicians running the assays require sufficient knowledge to manage DNA contamination, an issue with all nucleic acid amplification-based methods that needs to be addressed . Efforts to simplify DNA isolation methods , or reduce the requirement of electricity dependent equipment , have succeeded only on behalf of the sensitivity or specificity of the assay. In addition, the high cost of these commercially available assays has been put forward as a limiting factor in their uptake .
4.1. Lack of Specific Species Identification Is a Limitation with Currently Available Kits
Malaria species identification is important in areas where co-endemicity of non-falciparum species occurs, e.g., in areas where P. vivax infections—requiring additional liver-stage treatment—are prevalent . Although the Loopamp Pf and Pv detection kits provide specific identification of P. falciparum and P. vivax, it is at an additional cost of 5.20 euros per additional test and it still does not provide a very detailed and/or accurate description of the Plasmodium species composition . This has also shown to be of importance in areas where P. knowlesi—which has the ability to cause severe disease even at low parasite densities—is present. Molecular testing for quality assurance of microscopy-confirmed cases in Indonesia recently found that microscopy was unable to identify or miss-classified up to 56% of confirmed malaria cases, half of which were later determined to be P. knowlesi infections . Although the Loopamp Pan Detection kit successfully detected these infections in dried blood spots (unlike a more standard 18S rRNA nested PCR reference targeting the four human-only species), further species identification was limited by the unavailability of species-specific testing with the platform used. This highlights the difficulties of malaria species identification at the point-of-care and reference laboratory levels in settings where co-endemicity of non-falciparum species occurs .
4.2. The Loopamp Malaria Detection Kits Provide Several Advantages in Low Resource Settings
The lower cost of the Loopamp Malaria Detection Kits, together with the endorsement from FIND, is likely the main reason as to why primarily these kits have been used in malaria endemic settings . The simple and cheap boil and spin extraction method that can be used together with the Loopamp kits reduces the costs of extraction and can be performed near point-of-care reducing the time to result. A downside with the boil and spin method is that the extracted DNA is most likely not suitable for conventional PCR due to the presence of Taq polymerase inhibitors such as hemoglobin , and is not recommended for freezing due to the instability of the DNA . However, the Loopamp assay has successfully been conducted on finger prick blood samples collected in extraction buffer and stored at −80 °C for up to one year before the boil and spin extraction was performed .
An additional advantage with the Loopamp kits is the possibility of using dried blood spots on filter paper, as supposed to venous blood samples or whole blood from finger pricks . This allows easy transport and storage of samples at room temperature, allowing the assays to be conducted at a central laboratory rather than at point-of-care. Furthermore, if biological material is stored on filter papers, then LAMP positive samples can easily be transferred to more sophisticated laboratory facilities for more detailed analysis by PCR, e.g., for Plasmodium species identification and parasite density quantification.
4.3. Improving Throughput Will Aid Large Prevalence Surveys
Improved throughput for screening of larger numbers of samples simultaneously would benefit routine prevalence surveys in areas aiming at malaria elimination . High-throughput performance primarily depends on the DNA extraction capacity, which is considered the main bottleneck with large number of samples . The centrifugation-free methods available for both commercial kits have shown good clinical sensitivity, but at a greater cost per sample and they are not compatible with testing of large number of samples . The high throughput extraction platform developed for use with the Loopamp kits requires highly specific equipment , without significantly improving the time-to-results compared to boil and spin extraction of the same number of samples . Simplifying the sample processing protocol, e.g., by reducing the number of transfer steps, may increase throughput. This would also reduce the amount of plastic consumables needed, which is not only a benefit from an environmental point-of-view , but also given that sterile filter tips suitable for molecular assays are a commodity that are not readily available in low-resource-settings .
Finally, the use of the naked eye or a UV-lamp for the read-out of LAMP results is comparably less objective than the use of specialized equipment such as the Illumipro-10™ incubator or the LA-500 real-time turbidimeter . However, these instruments have limited capacity of 10 or 16 samples per run. Providing a more objective format for result readout on a larger scale could be a useful, especially for use in low prevalence areas where the occurrence of positive samples is rare and might be missed.
The entry is from 10.3390/diagnostics11020336
- Roth, J.M.; Korevaar, D.A.; Leeflang, M.M.; Mens, P.F. Molecular malaria diagnostics: A systematic review and meta-analysis. Crit. Rev. Clin. Lab. Sci. 2016, 53, 87–105.
- Notomi, T.; Okayama, H.; Masubuchi, H.; Yonekawa, T.; Watanabe, K.; Amino, N.; Hase, T. Loop-mediated isothermal amplification of DNA. Nucleic Acids Res. 2000, 28, E63.
- Oriero, E.C.; Jacobs, J.; Van Geertruyden, J.P.; Nwakanma, D.; D’Alessandro, U. Molecular-based isothermal tests for field diagnosis of malaria and their potential contribution to malaria elimination. J. Antimicrob. Chemother. 2015, 70, 2–13.
- Han, E.T. Loop-mediated isothermal amplification test for the molecular diagnosis of malaria. Expert Rev. Mol. Diagn. 2013, 13, 205–218.
- Abdul-Ghani, R.; Al-Mekhlafi, A.M.; Karanis, P. Loop-mediated isothermal amplification (lamp) for malarial parasites of humans: Would it come to clinical reality as a point-of-care test? Acta Trop. 2012, 122, 233–240.
- Polley, S.D.; Mori, Y.; Watson, J.; Perkins, M.D.; González, I.J.; Notomi, T.; Chiodini, P.L.; Sutherland, C.J. Mitochondrial DNA targets increase sensitivity of malaria detection using loop-mediated isothermal amplification. J. Clin. Microbiol. 2010, 48, 2866–2871.
- Han, E.T.; Watanabe, R.; Sattabongkot, J.; Khuntirat, B.; Sirichaisinthop, J.; Iriko, H.; Jin, L.; Takeo, S.; Tsuboi, T. Detection of four plasmodium species by genus- and species-specific loop-mediated isothermal amplification for clinical diagnosis. J. Clin. Microbiol. 2007, 45, 2521–2528.
- Lucchi, N.W.; Demas, A.; Narayanan, J.; Sumari, D.; Kabanywanyi, A.; Kachur, S.P.; Barnwell, J.W.; Udhayakumar, V. Real-time fluorescence loop mediated isothermal amplification for the diagnosis of malaria. PLoS ONE 2010, 5, e13733.
- Lucchi, N.W.; Ndiaye, D.; Britton, S.; Udhayakumar, V. Expanding the malaria molecular diagnostic options: Opportunities and challenges for loop-mediated isothermal amplification tests for malaria control and elimination. Expert Rev. Mol. Diagn. 2018, 18, 195–203.
- Poon, L.L.; Wong, B.W.; Ma, E.H.; Chan, K.H.; Chow, L.M.; Abeyewickreme, W.; Tangpukdee, N.; Yuen, K.Y.; Guan, Y.; Looareesuwan, S.; et al. Sensitive and inexpensive molecular test for falciparum malaria: Detecting plasmodium falciparum DNA directly from heat-treated blood by loop-mediated isothermal amplification. Clin. Chem. 2006, 52, 303–306.
- Zainabadi, K. Ultrasensitive diagnostics for low density asymptomatic plasmodium falciparum infections in low transmission settings. J. Clin. Microbiol. 2020.
- Mohon, A.N.; Getie, S.; Jahan, N.; Alam, M.S.; Pillai, D.R. Ultrasensitive loop mediated isothermal amplification (us-lamp) to detect malaria for elimination. Malar. J. 2019, 18, 350.
- Kemleu, S.; Guelig, D.; Eboumbou Moukoko, C.; Essangui, E.; Diesburg, S.; Mouliom, A.; Melingui, B.; Manga, J.; Donkeu, C.; Epote, A.; et al. A field-tailored reverse transcription loop-mediated isothermal assay for high sensitivity detection of plasmodium falciparum infections. PLoS ONE 2016, 11, e0165506.
- Zhang, Y.; Yao, Y.; Du, W.; Wu, K.; Xu, W.; Lin, M.; Tan, H.; Li, J. Development of loop-mediated isothermal amplification with plasmodium falciparum unique genes for molecular diagnosis of human malaria. Pathog. Glob. Health 2017, 111, 247–255.
- Oriero, C.E.; van Geertruyden, J.P.; Jacobs, J.; D’Alessandro, U.; Nwakanma, D. Validation of an apicoplast genome target for the detection of plasmodium species using polymerase chain reaction and loop mediated isothermal amplification. Clin. Microbiol. Infect. 2015, 21, 686.e1–686.e7.
- Port, J.R.; Nguetse, C.; Adukpo, S.; Velavan, T.P. A reliable and rapid method for molecular detection of malarial parasites using microwave irradiation and loop mediated isothermal amplification. Malar. J. 2014, 13, 454.
- Modak, S.S.; Barber, C.A.; Geva, E.; Abrams, W.R.; Malamud, D.; Ongagna, Y.S. Rapid point-of-care isothermal amplification assay for the detection of malaria without nucleic acid purification. Infect. Dis. 2016, 9, 1–9.
- Hayashida, K.; Kajino, K.; Simukoko, H.; Simuunza, M.; Ndebe, J.; Chota, A.; Namangala, B.; Sugimoto, C. Direct detection of falciparum and non-falciparum malaria DNA from a drop of blood with high sensitivity by the dried-lamp system. Parasites Vectors 2017, 10, 26.
- Serra-Casas, E.; Guetens, P.; Chiheb, D.; Gamboa, D.; Rosanas-Urgell, A. A pilot evaluation of alternative procedures to simplify lamp-based malaria diagnosis in field conditions. Acta Trop. 2019, 200, 105125.
- Mohon, A.N.; Elahi, R.; Khan, W.A.; Haque, R.; Sullivan, D.J., Jr.; Alam, M.S. A new visually improved and sensitive loop mediated isothermal amplification (lamp) for diagnosis of symptomatic falciparum malaria. Acta Trop. 2014, 134, 52–57.
- Lucchi, N.W.; Ljolje, D.; Silva-Flannery, L.; Udhayakumar, V. Use of malachite green-loop mediated isothermal amplification for detection of plasmodium spp. Parasites. PLoS ONE 2016, 11, e0151437.
- Singh, R.; Singh, D.P.; Savargaonkar, D.; Singh, O.P.; Bhatt, R.M.; Valecha, N. Evaluation of sybr green i based visual loop-mediated isothermal amplification (lamp) assay for genus and species-specific diagnosis of malaria in p. Vivax and p. Falciparum endemic regions. J. Vector Borne Dis. 2017, 54, 54–60.
- Yongkiettrakul, S.; Jaroenram, W.; Arunrut, N.; Chareanchim, W.; Pannengpetch, S.; Suebsing, R.; Kiatpathomchai, W.; Pornthanakasem, W.; Yuthavong, Y.; Kongkasuriyachai, D. Application of loop-mediated isothermal amplification assay combined with lateral flow dipstick for detection of plasmodium falciparum and plasmodium vivax. Parasitol. Int. 2014, 63, 777–784.
- Picot, S.; Cucherat, M.; Bienvenu, A.L. Systematic review and meta-analysis of diagnostic accuracy of loop-mediated isothermal amplification (lamp) methods compared with microscopy, polymerase chain reaction and rapid diagnostic tests for malaria diagnosis. Int. J. Infect. Dis. 2020, 98, 408–419.
- Selvarajah, D.; Naing, C.; Htet, N.H.; Mak, J.W. Loop-mediated isothermal amplification (lamp) test for diagnosis of uncomplicated malaria in endemic areas: A meta-analysis of diagnostic test accuracy. Malar. J. 2020, 19, 211.
- Nolasco, O.; Infante, B.; Contreras-Mancilla, J.; Incardona, S.; Ding, X.C.; Gamboa, D.; Torres, K. Diagnosis of plasmodium vivax by loop-mediated isothermal amplification in febrile patient samples from loreto, perú. Am. J. Trop. Med. Hyg. 2020, 103, 1549–1552.
- Aydin-Schmidt, B.; Xu, W.; González, I.J.; Polley, S.D.; Bell, D.; Shakely, D.; Msellem, M.I.; Björkman, A.; Mårtensson, A. Loop mediated isothermal amplification (lamp) accurately detects malaria DNA from filter paper blood samples of low density parasitaemias. PLoS ONE 2014, 9, e103905.
- Polley, S.D.; Bell, D.; Oliver, J.; Tully, F.; Perkins, M.D.; Chiodini, P.L.; González, I.J. The design and evaluation of a shaped filter collection device to sample and store defined volume dried blood spots from finger pricks. Malar. J. 2015, 14, 45.
- Vincent, J.P.; Komaki-Yasuda, K.; Iwagami, M.; Kawai, S.; Kano, S. Combination of pure-DNA extraction and lamp-DNA amplification methods for accurate malaria diagnosis on dried blood spots. Malar. J. 2018, 17, 373.
- De Koninck, A.S.; Cnops, L.; Hofmans, M.; Jacobs, J.; Van den Bossche, D.; Philippé, J. Diagnostic performance of the loop-mediated isothermal amplification (lamp) based illumigene(®) malaria assay in a non-endemic region. Malar. J. 2017, 16, 418.
- Lucchi, N.W.; Gaye, M.; Diallo, M.A.; Goldman, I.F.; Ljolje, D.; Deme, A.B.; Badiane, A.; Ndiaye, Y.D.; Barnwell, J.W.; Udhayakumar, V.; et al. Evaluation of the illumigene malaria lamp: A robust molecular diagnostic tool for malaria parasites. Sci. Rep. 2016, 6, 36808.
- Ponce, C.; Kaczorowski, F.; Perpoint, T.; Miailhes, P.; Sigal, A.; Javouhey, E.; Gillet, Y.; Jacquin, L.; Douplat, M.; Tazarourte, K.; et al. Diagnostic accuracy of loop-mediated isothermal amplification (lamp) for screening patients with imported malaria in a non-endemic setting. Parasite (Parisfrance) 2017, 24, 53.
- Hsiang, M.S.; Greenhouse, B.; Rosenthal, P.J. Point of care testing for malaria using lamp, loop mediated isothermal amplification. J. Infect. Dis. 2014, 210, 1167–1169.
- Rypien, C.; Chow, B.; Chan, W.W.; Church, D.L.; Pillai, D.R. Detection of plasmodium infection by the illumigene malaria assay compared to reference microscopy and real-time pcr. J. Clin. Microbiol. 2017, 55, 3037–3045.
- FIND. Manual of Standard Operating Procedures for Malaria Lamp. Available online: (accessed on 5 January 2021).
- Cuadros, J.; Martin Ramírez, A.; González, I.J.; Ding, X.C.; Perez Tanoira, R.; Rojo-Marcos, G.; Gómez-Herruz, P.; Rubio, J.M. Lamp kit for diagnosis of non-falciparum malaria in plasmodium ovale infected patients. Malar. J. 2017, 16, 20.
- Piera, K.A.; Aziz, A.; William, T.; Bell, D.; González, I.J.; Barber, B.E.; Anstey, N.M.; Grigg, M.J. Detection of plasmodium knowlesi, plasmodium falciparum and plasmodium vivax using loop-mediated isothermal amplification (lamp) in a co-endemic area in malaysia. Malar. J. 2017, 16, 29.
- Cook, J.; Aydin-Schmidt, B.; González, I.J.; Bell, D.; Edlund, E.; Nassor, M.H.; Msellem, M.; Ali, A.; Abass, A.K.; Mårtensson, A.; et al. Loop-mediated isothermal amplification (lamp) for point-of-care detection of asymptomatic low-density malaria parasite carriers in zanzibar. Malar. J. 2015, 14, 43.
- Hopkins, H.; González, I.J.; Polley, S.D.; Angutoko, P.; Ategeka, J.; Asiimwe, C.; Agaba, B.; Kyabayinze, D.J.; Sutherland, C.J.; Perkins, M.D.; et al. Highly sensitive detection of malaria parasitemia in a malaria-endemic setting: Performance of a new loop-mediated isothermal amplification kit in a remote clinic in uganda. J. Infect. Dis. 2013, 208, 645–652.
- Polley, S.D.; González, I.J.; Mohamed, D.; Daly, R.; Bowers, K.; Watson, J.; Mewse, E.; Armstrong, M.; Gray, C.; Perkins, M.D.; et al. Clinical evaluation of a loop-mediated amplification kit for diagnosis of imported malaria. J. Infect. Dis. 2013, 208, 637–644.
- Aydin-Schmidt, B.; Morris, U.; Ding, X.C.; Jovel, I.; Msellem, M.I.; Bergman, D.; Islam, A.; Ali, A.S.; Polley, S.; Gonzalez, I.J.; et al. Field evaluation of a high throughput loop mediated isothermal amplification test for the detection of asymptomatic plasmodium infections in zanzibar. PLoS ONE 2017, 12, e0169037.
- Morris, U.; Khamis, M.; Aydin-Schmidt, B.; Abass, A.K.; Msellem, M.I.; Nassor, M.H.; González, I.J.; Mårtensson, A.; Ali, A.S.; Björkman, A.; et al. Field deployment of loop-mediated isothermal amplification for centralized mass-screening of asymptomatic malaria in zanzibar: A pre-elimination setting. Malar. J. 2015, 14, 205.
- Perera, R.S.; Ding, X.C.; Tully, F.; Oliver, J.; Bright, N.; Bell, D.; Chiodini, P.L.; Gonzalez, I.J.; Polley, S.D. Development and clinical performance of high throughput loop-mediated isothermal amplification for detection of malaria. PLoS ONE 2017, 12, e0171126.
- Cuadros, J.; Pérez-Tanoira, R.; Prieto-Pérez, L.; Martin-Martin, I.; Berzosa, P.; González, V.; Tisiano, G.; Balcha, S.; Ramos, J.M.; Górgolas, M. Field evaluation of malaria microscopy, rapid malaria tests and loop-mediated isothermal amplification in a rural hospital in south western ethiopia. PLoS ONE 2015, 10, e0142842.
- Sema, M.; Alemu, A.; Bayih, A.G.; Getie, S.; Getnet, G.; Guelig, D.; Burton, R.; LaBarre, P.; Pillai, D.R. Evaluation of non-instrumented nucleic acid amplification by loop-mediated isothermal amplification (nina-lamp) for the diagnosis of malaria in northwest ethiopia. Malar. J. 2015, 14, 44.
- Katrak, S.; Murphy, M.; Nayebare, P.; Rek, J.; Smith, M.; Arinaitwe, E.; Nankabirwa, J.I.; Kamya, M.; Dorsey, G.; Rosenthal, P.J.; et al. Performance of loop-mediated isothermal amplification for the identification of submicroscopic plasmodium falciparum infection in uganda. Am. J. Trop. Med. Hyg. 2017, 97, 1777–1781.
- Serra-Casas, E.; Manrique, P.; Ding, X.C.; Carrasco-Escobar, G.; Alava, F.; Gave, A.; Rodriguez, H.; Contreras-Mancilla, J.; Rosas-Aguirre, A.; Speybroeck, N.; et al. Loop-mediated isothermal DNA amplification for asymptomatic malaria detection in challenging field settings: Technical performance and pilot implementation in the peruvian amazon. PLoS ONE 2017, 12, e0185742.
- Tambo, M.; Auala, J.R.; Sturrock, H.J.; Kleinschmidt, I.; Bock, R.; Smith, J.L.; Gosling, R.; Mumbengegwi, D.R. Evaluation of loop-mediated isothermal amplification as a surveillance tool for malaria in reactive case detection moving towards elimination. Malar. J. 2018, 17, 255.
- Mhamilawa, L.E.; Aydin-Schmidt, B.; Mmbando, B.P.; Ngasala, B.; Morris, U. Detection of plasmodium falciparum by light microscopy, loop-mediated isothermal amplification, and polymerase chain reaction on day 3 after initiation of artemether-lumefantrine treatment for uncomplicated malaria in bagamoyo district, tanzania: A comparative trial. Am. J. Trop. Med. Hyg. 2019, 101, 1144–1147.
- Björkman, A.; Morris, U. Why asymptomatic plasmodium falciparum infections are common in low-transmission settings. Trends Parasitol. 2020, 36, 898–905.
- Slater, H.C.; Ross, A.; Felger, I.; Hofmann, N.E.; Robinson, L.; Cook, J.; Goncalves, B.P.; Bjorkman, A.; Ouedraogo, A.L.; Morris, U.; et al. The temporal dynamics and infectiousness of subpatent plasmodium falciparum infections in relation to parasite density. Nat Commun 2019, 10, 1433.
- Hsiang, M.S.; Ntshalintshali, N.; Kang Dufour, M.S.; Dlamini, N.; Nhlabathi, N.; Vilakati, S.; Malambe, C.; Zulu, Z.; Maphalala, G.; Novotny, J.; et al. Active case finding for malaria: A 3-year national evaluation of optimal approaches to detect infections and hotspots through reactive case detection in the low-transmission setting of eswatini. Clin. Infect. Dis. 2020, 70, 1316–1325.
- Vallejo, A.F.; Martínez, N.L.; González, I.J.; Arévalo-Herrera, M.; Herrera, S. Evaluation of the loop mediated isothermal DNA amplification (lamp) kit for malaria diagnosis in p. Vivax endemic settings of colombia. Plos Negl. Trop. Dis. 2015, 9, e3453.
- WHO. Policy Brief on Malaria Diagnostics in Low-Transmission Settings; September 2014. Available online: (accessed on 29 January 2021).
- WHO. Mass Drug Administration, Mass Screening and Treatment and Focal Screening and Treatment for Malaria: Who Evidence Review Group Meeting Report; 16–18 September 2015. Available online: (accessed on 29 January 2021).
- Katrak, S.; Day, N.; Ssemmondo, E.; Kwarisiima, D.; Midekisa, A.; Greenhouse, B.; Kamya, M.; Havlir, D.; Dorsey, G. Community-wide prevalence of malaria parasitemia in hiv-infected and uninfected populations in a high-transmission setting in uganda. J. Infect. Dis. 2016, 213, 1971–1978.
- Nankabirwa, J.I.; Briggs, J.; Rek, J.; Arinaitwe, E.; Nayebare, P.; Katrak, S.; Staedke, S.G.; Rosenthal, P.J.; Rodriguez-Barraquer, I.; Kamya, M.R.; et al. Persistent parasitemia despite dramatic reduction in malaria incidence after 3 rounds of indoor residual spraying in tororo, uganda. J. Infect. Dis. 2019, 219, 1104–1111.
- Rek, J.; Katrak, S.; Obasi, H.; Nayebare, P.; Katureebe, A.; Kakande, E.; Arinaitwe, E.; Nankabirwa, J.I.; Jagannathan, P.; Drakeley, C.; et al. Characterizing microscopic and submicroscopic malaria parasitaemia at three sites with varied transmission intensity in uganda. Malar. J. 2016, 15, 470.
- Katrak, S.; Nayebare, P.; Rek, J.; Arinaitwe, E.; Nankabirwa, J.I.; Kamya, M.; Dorsey, G.; Rosenthal, P.J.; Greenhouse, B. Clinical consequences of submicroscopic malaria parasitaemia in uganda. Malar. J. 2018, 17, 67.
- Dlamini, N.; Hsiang, M.S.; Ntshalintshali, N.; Pindolia, D.; Allen, R.; Nhlabathi, N.; Novotny, J.; Kang Dufour, M.S.; Midekisa, A.; Gosling, R.; et al. Low-quality housing is associated with increased risk of malaria infection: A national population-based study from the low transmission setting of swaziland. Open Forum Infect. Dis. 2017, 4, ofx071.
- McCreesh, P.; Mumbengegwi, D.; Roberts, K.; Tambo, M.; Smith, J.; Whittemore, B.; Kelly, G.; Moe, C.; Murphy, M.; Chisenga, M.; et al. Subpatent malaria in a low transmission african setting: A cross-sectional study using rapid diagnostic testing (rdt) and loop-mediated isothermal amplification (lamp) from zambezi region, namibia. Malar. J. 2018, 17, 480.
- Smith, J.L.; Auala, J.; Tambo, M.; Haindongo, E.; Katokele, S.; Uusiku, P.; Gosling, R.; Kleinschmidt, I.; Mumbengegwi, D.; Sturrock, H.J.W. Spatial clustering of patent and sub-patent malaria infections in northern namibia: Implications for surveillance and response strategies for elimination. PLoS ONE 2017, 12, e0180845.
- Lufungulo Bahati, Y.; Delanghe, J.; Bisimwa Balaluka, G.; Sadiki Kishabongo, A.; Philippé, J. Asymptomatic submicroscopic plasmodium infection is highly prevalent and is associated with anemia in children younger than 5 years in south kivu/democratic republic of congo. Am. J. Trop. Med. Hyg. 2020, 102, 1048–1055.
- Goyal, K.; Kaur, H.; Sehgal, A.; Sehgal, R. Realamp loop-mediated isothermal amplification as a point-of-care test for diagnosis of malaria: Neither too close nor too far. J. Infect. Dis. 2015, 211, 1686.
- Paris, D.H.; Imwong, M.; Faiz, A.M.; Hasan, M.; Yunus, E.B.; Silamut, K.; Lee, S.J.; Day, N.P.; Dondorp, A.M. Loop-mediated isothermal pcr (lamp) for the diagnosis of falciparum malaria. Am. J. Trop. Med. Hyg. 2007, 77, 972–976.
- Coutrier, F.N.; Tirta, Y.K.; Cotter, C.; Zarlinda, I.; González, I.J.; Schwartz, A.; Maneh, C.; Marfurt, J.; Murphy, M.; Herdiana, H.; et al. Laboratory challenges of plasmodium species identification in aceh province, indonesia, a malaria elimination setting with newly discovered p. Knowlesi. PLoS Negl. Trop. Dis. 2018, 12, e0006924.
- Mohon, A.N.; Lee, L.D.; Bayih, A.G.; Folefoc, A.; Guelig, D.; Burton, R.A.; LaBarre, P.; Chan, W.; Meatherall, B.; Pillai, D.R. Nina-lamp compared to microscopy, rdt, and nested pcr for the detection of imported malaria. Diagn. Microbiol. Infect. Dis. 2016, 85, 149–153.
- Sattabongkot, J.; Tsuboi, T.; Han, E.T.; Bantuchai, S.; Buates, S. Loop-mediated isothermal amplification assay for rapid diagnosis of malaria infections in an area of endemicity in thailand. J. Clin. Microbiol. 2014, 52, 1471–1477.
- Please check and comment entries here.